mini hygro thermometer
Open the battery door and pull the battery insulator out, then remove the proteetion film
on the LCD, the digital thermometer is prepared for use now.
Minimum/maximum temperature and air humidity value
Press the button MAX/MIN once, The saved maximum values for indoor or temperature
and humidity will be displayed, after the values have been called, to delete maximum
values, press the button chear; Perss the button MAX/MIN once more,
The saved minimum values for temperature and humidity will be displayed,
to delete minimum values, press the button chear, After a short period of time the
device will display current temperature values automatically.
Adjusting the temperature unit(℃or℉)
Press the button (℃/℉)once, desired temperature unit ℃ (Celsius)or℉(degrees Fahrenheit).
Temperature:+/-1℃(1.8℉) within 0℃～35℃(32℉～95℉),
+/-2℃ at the remaining range
Humidity : 5%R.H,within 40%～80%R.H,and 0℃～35℃(32℉～95℉),5% at the remaining range
Unit size: 53(L) ×34(W) ×16(T)mm
Battery: AG13 or.LR44
Open the battery door and pull the battery insulator out, then remove the proteetion
film on the LCD, the digital thermometer is prepared for use now.
Left display R.H humidity and right display temperature, The temperature and humidity
sensors are both in sensor probe, Do not put the sensor probe into oil or the other liquid
When the display becomes to faint, the battery need to be replaced to assure accuracy.
Reset the battery if any problem.
Dispose of waste battery carefully.
The quality of instrument will not be ensured if the damage is caused by negligence,
improper usage or dimantling of the parts.